Expands data between Cursors A and B when the scroll is stopped during capture,
or during replay.
* The expansion in the Y-axis direction is not available.
Y Axis Operation
Click this button to revert the values set in the Y axis span and position to the
default values. The default values are the same values as those of when switching
Click this button to display plot marks at the actual sample points on the
Change the line width of the waveforms.(1/2/3/4/5)
* The line may be thicker than the selected number of dots due to circumstances regarding input signals.
Calculation CH On/
Use these buttons to set calculations 1-4 to On/Off. On : Calculation results are
shown as waveforms and digital values. Off : Do not perform calculations.The
calculation results are only shown in Y-T display, and do not affect the captured
Sets the calculation channel for which calculation is to be set
(* The unit conforms to CH
Use this button to set the variable for a linear expression between channels.
A • CH X * B • CH Y + C
The expression you set appears at the bottom of this window.
(A and B are arbitrary coefﬁcients. * is an arithmetic operation (+-x÷). X and Y are arbitrary channels, and
C is an arbitrary constant.)
Displays the calculation speciﬁed in the Expression setting.
Click this icon to input a comment above the waveform of the desired channel during a data capture (replay) operation.
If the scroll is ON, the input position is at the "Comment Input" in the upper part of a waveform. If the scroll is OFF, it is
the position of Cursor A.
Select a channel for entering a comment.
Enters a comment. Up to 20 comments can be entered. If you change a position
where a comment has already been entered, the entered comment will also be
Click this button to input the comment..
Click this button to delete the comment that was input
Comments will be displayed based on the scale speciﬁed at the start of the data capture
operation.If the Y-axis scale is changed during data capture, the input comments will be
off positioned when displayed on the replay screen. To display the comments above the
waveform, change the Y-axis scale after the replay.