Operating Manual 1303.6580.32-05
Numeric suffix, not used in this command. Implemented for compatibility with
OFF – Independent color scheme in new diagram area. Moved traces change
ON – Color scheme in new diagram area continues the previous color scheme.
Moved traces keep their color.
*RST does not affect the color settings; see also description of the Preset
Device-specific, command or query
*RST; DISP:CMAP13:RGB 1,0,0
Create diagram area no. 1 (with default trace showing the S-parameter S
color the trace red.
DISP:CMAP:TRAC:COL OFF; DISP:WIND2:STAT ON
Select independent color schemes for new diagram areas. Create a new
diagram area no. 2.
CALC:PAR:SDEF 'TRC2', 'S11'; DISP:WIND2:TRAC2:FEED 'TRC2'
Create a new trace named TRC2 and display the trace in a new diagram area
no. 2. The new trace is red like the first trace.
DISP:CMAP:TRAC:COL ON; DISP:WIND3:STAT ON
Continue the same color scheme in new diagram areas. Create a new diagram
area no. 3.
CALC:PAR:SDEF 'TRC3', 'S22'; DISP:WIND3:TRAC3:FEED 'TRC3'
Create a new trace named TRC3 and display the trace in a new diagram area
no. 3. The new trace is not red.
Executes the function of a key with a specified name and switches to the local screen.
You can use this command to execute part of your measurement task manually; see Combining
Manual and Remote Control. Menu keys that initiate events are executed immediately, because no
additional input is necessary. However, DISPlay:MENU:KEY:EXECute can be used as well for manual
entries in numeric entry bars, dialogs or wizards.
Name of the key as shown in the analyzer's softkey bar (case-sensitive string
variable, may contain blanks but no dots). If the name is not unique, the complete
menu path can be specified, e.g. 'Trace:Format:Phase'. Menus, submenus and
menu function must be separated by colons.
Device-specific, no query
*RST; DISP:MENU:KEY:EXEC 'S11'; EXEC 'Trace:Format:Phase'