Be sure to agree to the terms of the software license be-
e) must be connected to the host PC before FluorEs-
fore using the software.
A USB dongle is supplied with FluorEssence™. This dongle
sence™ will operate.
Copying, disassembly, or removal of the softkey is illegal.
e CPU, monitor, keyboard, mouse, speakers, printers, etc.
he spectrofluorometer system is controlled by FluorEssence™ spectroscopy
tific, perform the installa-
on. Contact a HORIBA Scientific Sales Representative for recommended spe-
Before the FluorEssence™ software can be installed, however, Windows
be installed already and operating properly. Refer to the Windows
the host computer for software security. The softkey should be left in place on the host
computer at all times.
sed from HORIBA Scientific or another supplier.
Set up the host PC reasonably close to the FluoroMax -4 system. The
limitation is the length of the USB cable. The recommended location fo
the PC is just to the right of the spectrofluorometer, but other p
Follow the instructions for the host PC to set up the computer system,
Connect the FluoroMax
to the computer.
Attach the free end of the USB cable to a USB receptacle on t
With all devices OFF, plug the power cords from the m
, and the printer into properly grounded re
Install any accessories that arrived with the system, using the
instructions that accompany the accessories.
See Chapter 12 for a detailed list of accessories.
Install the FluorEssence™ software.
software operating within the Windows
environment. If the computer and
software were purchased from HORIBA Scientific, the software installation is
complete. If the computer is not from HORIBA Scien
cifications for a suitable host computer.
came with the computer for installation instructions.
rEssence™ software is supplied on one CD-ROM. Follow the
™ User’s Guide
for details on installation.
utside of the USA:
tside of the USA receive a softkey device that connects to the printer p